PIB 37-333 New Instrument Software LICO 6xx and TU5200
Global; Hach Lab; 1/31/2019
Level One - Customers-Distributors-Internal
A new Software update is available on the Hach website for the LICO 6xx spectrophotometers and the TU5200 Benchtop Turbidimeters.
We recommend that customers should contact the Hach representative or authorized dealer from whom the device was purchased. There we offer an instrument inspection or repair and to update the instrument to the most recent software version.
The LICO software update to V1.10 contains the following changes:
A sample ID will be now deleted after removing the sample vial/cuvette out of the cell compartment (GLP-reason, workflow)
A linefeed is added to a Thermal printer print-out at the end of the print to allow a direct paper cut w/o pressing LF (workflow improved)
ADMI color number determination: The upper range is increased to 1000 ADMI
Auto-Print-out function now selectable for all type of printers (before it was only for thermal printer DPU-S445 selectable)
Bugfix: Change of Addista color Lot-number in AQA menu caused a crash
Bugfix: Unused Claros-Service-Codes deleted
The file "lcp6.tar.gz" updates the operating system of the LICO 6xx/LCx IV to the most recent version. Please DO NOT update your instrument if you don´t want to have the new features implemented. A downgrade to an older/prior software version is not possible.
The TU5200 software update to V1.06 contains the following changes:
Registration Network Share using Domain/Username (>= Kernel #9)
New revision of user manuals implemented
Bugfix: RapidCal error after factory calibration
The file " tu5200.tar.gz" updates the operating system of the TU5200 to the most recent version. Please DO NOT update your instrument if you don´t want to have the new features implemented. A downgrade to an older/prior software version is not possible.
All software updates can be downloaded from www.hach.com.